Location : 4th Floor Serial Number: AD0510243 Year of Manufacture: 2005 To Include: Assistant Chairs x 2 Belmont Clesta Spittoon – Serial Number: FB05G0204 Belmont IO5000 501T Dental Light – Serial Number: CG050035 Water/Air Footswitch ***Our Engineer's Comments:The dental suite powers up ok and the electric chair is fully functional, with all the motorised positions working fine. The upholstery is in good condition. The dental light is functional with its three (high, medium and low) light intensities functional. The spittoon appears to be in good physical and cosmetic condition, with ejector hose and triple syringe in place and functional. The dental instrument cart appears to be in good condition, with all the couplings and tubing in place. The triple syringe is included, but the tips are not included. Water container is missing. All the tubing, stop valves x2 and electrical connectors are in place and intact. The unit has been tested with water and air supply. Includes two dental chairs which are in average condition as there is some physical damage. The air/water footswitch is exposing its tubing’s near the footswitch end. Overall, the unit is in good physical and cosmetic condition but the cover for the water gauge is missing on the flow box. Please note: ALL items will be disconnected by Hilditch and made ready for collection for the buyer. Items will not be packed or palletised; all buyers will need to arrive with a suitable vehicle with Tail Lift. There are no lifting facilities on site. Disconnection costs will be covered by Hilditch, but all other charges will be at the buyer’s expense.
All lots are sold as seen. Any views expressed are for guidance only - please refer to our conditions of sale. It is the responsibility of the purchaser or their agent to arrange for the safe removal of these lots. Any damage to buildings is the responsibility of the Purchaser & must be made good and safe and the entire area left clean and tidy.Printable Lot Details